Read or Download Agilent 16533A, 16534A Digitizing Oscilloscope (help volume) PDF
Best nonfiction_6 books
This infrequent set of prints is nearly mythical between students of the army uniform. those prints have been made up of real watercolor work of the themes in Turkey through an itinerant French painter named de Molleville. One wonders if he have been maybe a undercover agent despatched through Napoleon to evaluate the army may well of Turkey?
- Model 1903 Springfield RiflesNational Rifle Association of AmericaASIN: B001KLFB66
- Dzog Chen and Zen
- The Jewel Ladder:A Preliminary Nyingma Lamrim
- Food Services (Ferguson Career Launcher)
- Narsinh Metha
Extra resources for Agilent 16533A, 16534A Digitizing Oscilloscope (help volume)
In the menu bar, select Setup. 2. Select Display... The Display Setup dialog appears. 3. Under Colors, select the channel to modify. 4. Select the color you want it to be. 5. Select the Edit Colors... button to change a color's value. NOTE: If you Close the Color Edit box, the new color values will be used in this session only. If you Apply the color values, they will be used in this session and following sessions. To restore the factory colors, select the Reset Defaults button. Changing the Grid The Agilent Technologies 16533A or 16534A Digitizing Oscilloscope has a 10 by 8 display graticule grid which you can turn on or off.
4. In the Correlation Error dialog that appears, select Group Run for both instruments. 5. Select the Group Run or Run All button to start the acquisition. 36 Chapter 1: Agilent Technologies 16533/34A Digitizing Oscilloscope Combining the Oscilloscope with a Logic Analyzer 6. To view the waveforms together, open the display tool. • For a Waveform display, select one of the labels and choose Insert before... or Insert after.... In the Label Dialog, select the label you want to insert, then select the Apply button.
The value is calculated by summing all the data points on the screen and dividing by the number of them. Period Period is defined as the time between the 50% threshold crossings of two consecutive, like-polarity edges. The Agilent Technologies 16533A or 16534A Digitizing Oscilloscope starts the measurement at the leftmost edge of the display. Rise Time Rise time is defined as the time at the 90% threshold minus the time at the 10% threshold on the edge you are measuring. 63 Chapter 1: Agilent Technologies 16533/34A Digitizing Oscilloscope Automatic Measurements and Algorithms The Agilent Technologies 16533A or 16534A Digitizing Oscilloscope starts the measurement at the leftmost edge of the display.
Agilent 16533A, 16534A Digitizing Oscilloscope (help volume)